WirelessHART compliance verification receiver available
In a move advance development and compliance testing of WirelessHART devices, the HART Communication Foundation released the Wi-Analys Compliance Verification Receiver.
Austin, TX – In a move critical to the development and compliance testing of WirelessHART devices, the HART Communication Foundation , released the Wi-Analys Compliance Verification Receiver. The unit allows companies designing new wirelessly enabled HART devices to test and analyze their radio transmissions.
“The HART Wi-Analys compliance verification receiver is designed to capture all 802.15.4 transmissions operating in the 2.4 GHz frequency range, but focuses on those from WirelessHART devices,” says Ed Ladd, HCF director of technology programs. “The receiver has the capability of capturing data at up to 1,000 messages per second.”
Wally Pratt, chief engineer in the organization’s test laboratory, demonstrated the prototype receiver for Control Engineering. Significantly, the unit captures all transmissions on all 16 IEEE 802.11.4 channels from the unit under test (UUT), along with a time stamp precise to 1μs. Test analyst software then compares those messages with what the standard says should have been sent based on the test stimuli. It then logs the results into a searchable database. Engineers can then mine that data to help the UUT vendor find and fix problems before the UUT becomes a product inthe field.
“We’re really on the vendor’s side,” Pratt points out. “We help them make sure their products will be robust in the field.”
The test specification (test plan) for a Wireless HART device may include as many as 2,000,000 messages, and take several weeks to complete. Among that large number of messages, no more than 20 misses are allowed. “We may repeat messages hundreds of times to make sure the UUT gets it right every time,” Pratt continues.
The prototype receiver has been undergoing final development tests in the organization’s Austin, TX test laboratory, and has already proven invaluable in validating compliance of initial prototype devices sent to the Foundation for device registration. The Wi-Analys receiver has passed all required FCC and CE tests and is now available to verify HART compliance.
HCF describes itself as international, not-for-profit, membership organization supported by more than 200 companies worldwide.
Also read a story that includes a link to an audio clip: Listen in: First WirelessHART devices about to hit the market and What process control applications are OK for wireless? Learn from others to help your industrial wireless application .
– C.G. Masi , senior editor
Control Engineering News Desk
Register here and scroll down to select your choice of eNewsletters free .
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
2012 Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.