Tutorial: Vortex shedding flowmeters
One technology that scored well in Control Engineering’s August 2008 Product Research survey on flowmeters is vortex shedding. Vortex shedding flowmeters calculate liquid, gas, or steam flow by observing the vortices formed when a fluid stream passes an obstruction with known characteristics. When a bar (called a bluff body) is inserted across the interior of a pipe, fluid flowing throug...
One technology that scored well in Control Engineering’s August 2008 Product Research survey on flowmeters is vortex shedding.
Vortex shedding flowmeters calculate liquid, gas, or steam flow by observing the vortices formed when a fluid stream passes an obstruction with known characteristics.
When a bar (called a bluff body) is inserted across the interior of a pipe, fluid flowing through the pipe has to pass on one side or the other. As it passes, vortices form on both sides of the body. The frequency of these vortices is determined by the velocity of the fluid stream. Instrumentation devices use a detector to count the frequency, such as strain gages, ultrasonic, capacitance, magnetic, or other types of sensors to generate electrical pulses from the liquid vibrations. The transmitter counts these pulses and converts the reading to a stream velocity from which it calculates fluid volume.
This approach has many advantages:
Accuracy and turn-down ratio are sufficient for most general process applications;
Wide range of sizes available;
Durable and not prone to drift;
Stable, so they need little calibration or maintenance;
Good for gas, steam, and liquid;
Most can mount in any position so long as the pipe is full;
No moving parts;
Works with conductive or non-conductive liquids; and
Configuration options include spool or wafer designs, and insertion styles for larger sizes.
However, there are some downsides you should keep in mind:
Any design that introduces an obstruction causes a pressure drop and creates clogging potential. The extent varies between manufacturers;
Solids depositing on the bluff bar can disrupt accurate readings;
Two-phase flows and slurries are not well-suited for this kind of flowmeter;
Over time, the bluff bar’s shape may change from wear which can cause drift;
Liquid viscosity is generally limited to 30 cp;
Specific designs can be dedicated either to gas or liquid, but not both; and
Ensure sufficient up- and down-stream straight pipe to control turbulence.
Some of the more sophisticated designs add sensors for variables such as fluid temperature and pressure. These readings are often very useful, and can be used to calculate a mass flow reading when known fluid characteristics are added.
To read the Product Research article on flowmeters, visit controleng.com , click on the magazine tab at top of page, select “archive,” and visit the August 2008 issue. The full report is published in the Resource Center at controleng.com. It can be accessed by selecting the “Resource Center” tab at the top of the page.
- Events & Awards
- Magazine Archives
- Oil & Gas Engineering
- Salary Survey
- Digital Reports
Annual Salary Survey
Before the calendar turned, 2016 already had the makings of a pivotal year for manufacturing, and for the world.
There were the big events for the year, including the United States as Partner Country at Hannover Messe in April and the 2016 International Manufacturing Technology Show in Chicago in September. There's also the matter of the U.S. presidential elections in November, which promise to shape policy in manufacturing for years to come.
But the year started with global economic turmoil, as a slowdown in Chinese manufacturing triggered a worldwide stock hiccup that sent values plummeting. The continued plunge in world oil prices has resulted in a slowdown in exploration and, by extension, the manufacture of exploration equipment.
Read more: 2015 Salary Survey