Multivariable transmitters for flow, level measurements
ABB's 266 Multivariable transmitters are designed to directly calculate the mass flow or standard volume flow for gases, vapors and liquids and can correct parameters for discharge coefficient and the thermal expansion of pipelines.
ABB's 266 Multivariable transmitters are designed to directly calculate the mass flow or standard volume flow for gases, vapors and liquids with dynamic flow compensation from three process variables: Differential pressure, absolute pressure and temperature.
The transmitters enhance differential pressure flow measurements of DP primary elements such as Orifice Plates, Pitot Tubes, Venturis, and Wedge Meters. Apart from pressure dependent and/or temperature dependent changes to the density of the fluid, the transmitters also correct parameters as the discharge coefficient, thermal expansion of pipelines and primary device, along with the Reynolds number. The 266 Multivariable also offers the advantage that a single device replaces a standard DP transmitter, gauge transmitter, RTD, associated wiring and flow calculator.
In addition, these instruments can measure the level of liquid-filled tanks and boiler drum level with temperature and/or pressure dependent density changes of the process medium. With the inclusion of a tank shape specification, they can also measure liquid volume.
The 266 Multivariable offers the ABB exclusive TTG (Through The Glass) technology. This technology allows configuration of the instrument through the glass of the housing cover via four buttons on the display of the transmitter.
The transmitters contain a standard HART diagnostic called PILD (Plugged Impulse Line Diagnostics) detection. PILD monitors the impulse lines for clogging, identifying a possible fault before the measuring point fails, and signals this fault in a status message both locally and via HART. The 266 Multivariable transmitters also have pulse or frequency outputs for activating external totalizers or for signalling limit-value violation.
- See more Control Engineering sensor products.
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
Annual Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.