Multi-parameter level sensor adds measuring capabilities
E+H Levelflex FMP55 level transmitter uses both capacitance and guided wave radar technologies to measure the level and interface between two liquids.
Endress+Hauser has released its Levelflex FMP55 multiparameter level transmitter that can measure both the overall level of a liquid and the interface between two liquids. The company says the FMP55 is the first transmitter to use two level technologies built into one probe. The transmitter will switch between the two methods automatically in order to provide a concise and reliable interface measurement, even in the presence of emulsion or rag layers. The use of two technologies eliminates many problems associated with interface level applications.
Capacitance is not affected by emulsion or rag layers in interface applications, and the radar allows the transmitter to monitor the overall level as well as the interface level. The transmitter has a multi-echo tracking algorithm that can track up to 20 tank echoes simultaneously as the tank level rises or lowers. This ensures that the transmitter will not make a "jump" to an echo that is not the true level, such as a internal tank obstruction or heater coils. The level sensor has a measuring range of up to 33 ft (10 m). The transmitter offers a 4-20mA with HART output signal plus an optional second 4-20 mA output.
With its coated multiparameter probe, the FMP55 is well suited for interface measurement in the oil & gas and chemical industries. It can be used in process temperatures from -58 to 392 °F (-50 to 200 °C) and in pressures from -14.5 to 580 psi (-1 to 40 bar). The FMP55 carries a SIL 2 rating as certified by TÜV per IEC 61508 and has ATEX, IEC Ex, FM and CSA area classification approvals available.
The unit is equipped with a HistoROM built into the main housing of the transmitter. All of the setup parameters are stored on the HistoROM and, in the event of an electronics change-out, all of the information will be uploaded directly into the new electronics module. In the case of multiple units being set up at the same time with the same measurement parameters, the same HistoROM information can be transferred into a display that can be removed and installed in the next transmitter. This shortens setup time during field commissioning.
Visit the Control Engineering Process Control Channel.
Edited by Peter Welander, pwlenader(at)cfemedia.com
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
Annual Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.