Magnetostrictive level sensor for very deep tanks
MTS Systems has upped the maximum order length of its Level Plus MG level transmitter to 866 in. (22 m) without changing its level of accuracy.
MTS Systems Corp ., Sensors Division has increased the maximum order length of the Level Plus model MG level transmitter to 866 in. (22 m), which is approximately 20% longer than the previous available length. The increase in length does not affect the accuracy of the transmitter, which provides maximum non-linearity of 0.062 in. This level of accuracy is typically required for critical inventory and custody transfer applications.
The MG level transmitter's magnetostrictive design uses a moving float to provide digital output in the liquid-level marketplace for vessels from 20 in. (.508 m) to 866 in. (22 m). Outputs include Modbus, DDA (proprietary ASCII protocol), and Foundation Fieldbus. MTS says the MG transmitter features 3-in-1 measurement of the product level, interface level, and the average temperature from up to 12 temperature points, using a single process opening, eliminating the need for separate level and temperature transmitters.
Users can obtain direct volume readings by using a 100-point strap table that the user loads into the transmitter. By incorporating the strap table into the transmitter the user does not need multiple copies of the strap table or to code the additional formulas.
– Edited by Renee Robbins , senior editor
Control Engineering News Desk
Register here and scroll down to select your choice of eNewsletters free.
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
Annual Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.