Listen in: Ultrasonic technology offers versatility, economy
New electronic capabilities make for economy, simpler operation.
New family of ultrasonic flowmeters offers a useful combination of application versatility, simple operation, reliability, and economy for many general process applications. ( Click here to hear comments from Don Tyler, president of Clark Solutions .)
Clark Solutions ClarkSonic ultrasonic flowmeters are reportedly well suited for measuring flow in most clean liquids and many liquids with entrained solids. Applications include monitoring flow rates of chilled or heated water for HVAC installations, as well as in municipal, process, and industrial systems. The company says the family frees users from having to calibrate the transmitter to fluid temperature, viscosity, or density, making them very easy to deploy and very reliable. Output is standard 4-20 mA with user definable flow units.
ClarkSonic family has two main operating ranges. Model CSLFC ultrasonic transmitters are designed for flows to 400 GPM in
The company says the heart of the ClarkSonic flowmeters is a proprietary mixed signal ASIC, (application specific integrated circuit,) which allows sophisticated timing, control and transducer driver circuitry to be combined on a single integrated circuit. The ASIC uses a special algorithm that improves on standard single-path measurement techniques. ClarkSonic transmitters use the sing around method, where the ultrasonic transducer alternates between transmitting and receiving to measure differences in flight time between upstream and downstream transmissions. The flight time of the sound pulse from the transmitting transducer to the receiving transducer will be shortened if the pulse is launched in the direction of flow, and increased if launched opposite to the direction of flow. By alternating the transmitting and receiving transducers, the difference in these transit times can be used to calculate the velocity of the flow, which when multiplied by the area of the pipe results in a measure of volumetric flow rate. The primary advantage of the sing around method is that the velocity of the fluid is being measured independent of the relative speed of sound in that given fluid. That way, the output is unaffected by changes in fluid temperature, density and viscosity.
—Edited by Peter Welander, process industries editor, PWelander@cfemedia.com ,
Process Instrumentation & Sensors Monthly
Register here and scroll down to select your choice of free eNewsletters .
- Events & Awards
- Magazine Archives
- Oil & Gas Engineering
- Salary Survey
- Digital Reports
Annual Salary Survey
Before the calendar turned, 2016 already had the makings of a pivotal year for manufacturing, and for the world.
There were the big events for the year, including the United States as Partner Country at Hannover Messe in April and the 2016 International Manufacturing Technology Show in Chicago in September. There's also the matter of the U.S. presidential elections in November, which promise to shape policy in manufacturing for years to come.
But the year started with global economic turmoil, as a slowdown in Chinese manufacturing triggered a worldwide stock hiccup that sent values plummeting. The continued plunge in world oil prices has resulted in a slowdown in exploration and, by extension, the manufacture of exploration equipment.
Read more: 2015 Salary Survey