Level switch: Two-wire, loop-power device for tank monitoring, interface control
Can often replace mechanical flow and paddle switches without wiring changes.
Rugged NuTec LS2000 level switch from Fluid Components International (FCI) provides reliable level monitoring and interface control in process applications.
The LS2000 level switch is designed for two-wire, 4 to 20 mA loop power that reportedly allows the device to serve as a drop-in replacement for other two-wire mechanical float and paddle switches without any wiring modifications. This simplifies installation and reduces costs during plant retrofit projects. It is also available for 115/230 Vac or 24 Vdc input power and a 6A relay output.
The LS2000 level switch features a thermal mass flow sensor with no moving parts constructed with either 304/305 stainless steel or with all welded 316L stainless steel wetted surfaces. The device can measure level within +0.25 in. (6mm) from its element tip when top mounted. The side mount configuration is accurate to +0.125 (3mm). It is temperature compensated for process and plant environment swings of up to 100 °F (38 °C) to provide measurement accuracy and stability under harsh conditions.
Its thermal mass flow sensing technology reportedly makes it suitable for detecting and controlling interface in tanks and separation vessels. It can identify the interface of multiple media, such as liquids and foams, liquids and sediments, and liquids and resins.
The aluminum enclosure is rated NEMA 4X (IP66) for harsh environments. Approvals and ratings include: Class I and II, Div 1 and 2, Groups C and D, as well as FM, CSA, CE marking and ATEX (pending).
—Edited by Peter Welander, process industries editor, PWelander@cfemedia.com ,
Control Engineering Process Instrumentation & Sensors Monthly
Register here and scroll down to select your choice of free eNewsletters .
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
Annual Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.