High output accelerometer for machinery health monitoring
The 786-500-D2 accelerometer by Meggitt is designed for condition monitoring for rotating equipment and can be used in hazardous areas because of its Class I Div 2/Zone 2 certification.
Meggitt's 786-500-D2 is a new high output accelerometer for machinery health monitoring. The 786-500-D2 accelerometer carries CSA approvals for Class I Div 2/Zone 2 areas and has a high 500 mV/g output sensitivity and a broadband frequency response for multi-use applications.
The 786-500-D2 accelerometer is well suited for condition monitoring of a variety of rotating equipment including those in hazardous areas. Slow speed applications, such as wind turbine generators will benefit from the strong 500 mV/g output of the 786-500-D2 because it overcomes data collector noise that can mask peaks at low frequencies. In high speed applications, such as machine tool spindles or gear mesh frequencies the sensor’s higher sensitivity results in better detection of low level vibrations.
The accelerometer houses a 500 mV/g sensing element with a tight 5% sensitivity tolerance coupled with a ± 3 dB frequency response range that covers 0.2 to 14,000 Hz. As a multi purpose sensor with a full scale acceleration range of 10 g and an excellent broadband noise floor of 250 μg, the 786-500-D2 is ideal for applications where both ends of the machine spectrum need to be monitored simultaneously.
Class I Div 2/Zone 2 certification means the 786-500-D2 can be used in higher risk areas, those defined as areas where hazardous gas, vapors or mist would only be present under abnormal conditions. While not as stringent as Division 1/Zone 1 certifications, this CSA certified unit can be safely used on motors, pumps, fans and other process equipment located in hazardous areas. Since no safety barriers are required in Division 2/Zone 2 areas, installation of the sensor is simplified.
- Edited by Chris Vavra, Control Engineering, www.controleng.com
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
2012 Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.