Greenhouse gas flow meter for process, plant flue gas applications
A mass flow meter designed to monitor greenhouse gases can help users in chemical, food processing, petroleum refining, pharmaceutical, electric power generation, and other plants be in compliance with the EPA emission requirements.
Process and plant engineers who are concerned about the U.S. EPA’s greenhouse gas monitoring requirements will find the ST51 GHG mass flow meter from Fluid Components International (FCI) provides just what they need. The ST51 features a no-moving-parts design that’s non-clogging, and it operates over a wide flow range with low-flow sensitivity. It’s packaged in an explosion-proof transmitter, and the calibration is matched to the user’s actual gas composition and installation conditions. Flue gases, the mixed-composition hydrocarbon greenhouse gases that are the by-product of an industrial plant combustion process, can include nitrogen, carbon dioxide, oxygen, water vapor, sulfur oxides, nitrogen oxide, carbon monoxide, particulates, ozone, and methane.
The ST5 features a thermal mass, insertion-style flow element with flow accuracy to ±1% of reading over a broad flow range from 0.3 to 400 SFPS (0.08 to 122 MPS), and repeatability of ±0.5% of reading. The flow element is available for use in line sizes from 2 to 24 in. (51 to 610 mm). It operates over a wide turndown range of 100:1 and at temperatures from 0 to 250 °F (-18 to 121 °C). It withstands pressures up to 500 psig (34 bar).
The unit's robust thermal mass flow sensing element has no moving parts and no orifices to clog or foul, allowing it to attain virtually maintenance-free service in wet, dirty biogas applications. The flow element is constructed with a 316L stainless steel body and Hastelloy C-22 thermowell sensors to resist corrosion. It includes built-in temperature compensation circuitry for accurate, repeatable measurement as temperatures rise and fall seasonally.
Dual 4-20 mA analog outputs are user-assignable to flow rate and/or temperature, and there is a 0-1 kHz pulse output for totalized flow. The transmitter’s digital communications include an RS-232C port, and with units that have the digital display option there is a wireless IR link for PDA use. An optional integral digital display features a two-line, 16-character LCD screen that can be rotated in 90° increments for optimum viewing. The LCD screen’s top line is assigned to flow rate, and the bottom line is user-assignable for temperature readings or totalized flow.
Fluid Components International
- Edited by Renee R. Bassett for Control Engineering, www.controleng.com
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
2012 Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.