Application: Water utility brings calibration in house
Calibration--Do it yourself or farm it out?
The water department had two choices: Either keep paying a service contractor to calibrate flowmeters, or buy the same tool he used to do the job themselves. Their choice paid off.
After replacing aging turbine flowmeters with ABB magnetic flowmeters , the City of Mesa, AZ, water division had 36 new 12 inch devices to add to the calibration schedule. They had been calling an ABB service technician to calibrate other magnetic flowmeters, so the choice was either to expand the service contract or buy the necessary calibration tools and do all the magnetic units themselves.
Since those flowmeters are responsible for watching over 30 billion gallons of treated water annually, accuracy is critical and needs to stay around
The maintenance team purchased a CalMaster calibration verification tool. It is a portable, PC-based system designed for use with MagMaster brand magnetic flowmeters offered by ABB Instrumentation. The purchase included factory training of a division staff member so they gained in-house capability to verify and re-certify the calibrated accuracies not only of the new well meters but also of existing MagMaster flowmeters distributed throughout the system.
Each MagMaster primary sensor and its transmitter are factory-calibrated as a matched pair. The two are shipped with a memory card that contains a "fingerprint" with all pertinent calibration data for this specific sensor, including the shielded cable that carries the small millivolt measuring signal from the sensor to its transmitter.
The primary sensor stays in operation on the water line and connects by shielded cable to its stand-mounted flow transmitter. The continuous flow measurement is never interrupted. The PC contains software to carry out a 10- to 14-minute, preprogrammed verification procedure based on the data contained on the memory card. When the operator initiates the test, the PC carries out a check of the entire system: meter primary sensor, transmitter, and connecting cable.
To confirm that the test results are within set standards, the test operator reviews them on the display. If the results are positive, a certificate of validation for the tested metering system can be printed out and filed in the maintenance records.
The CalMaster also provides valuable diagnostic tools for preventive maintenance. It stores a history of calibration data from successive tests, so that trend curves for selected system components can be plotted on the PC. Thus, a gradual degradation in a component, such as a magmeter coil, can be detected and the component scheduled for repair or replacement.
—Edited by Peter Welander, process industries editor, PWelander@cfemedia.com ,
Process Instrumentation & Sensors Monthly
Register here and scroll down to select your choice of free eNewsletters .
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
Annual Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.