Miniature LVDT position sensors provide accurate position feedback
Macro Sensors CD 375 Series is available in full-scale measurement ranges from ±0.025-inch to ±2.0-inch, offering a core-to-bore radial clearance of 0.012-inch, repeatability with error of <0.01% of FSO.
The compact size and accurate output of Macro Sensors’ CD 375 Series of miniature LVDTs make the linear position sensors ideal for providing displacement feedback in different robotic applications.
Robots and robotic arms rely on sensors to determine range of motion or force applied to an object. Attached to a robotic arm, the Macro Sensors miniature LVDT linear position sensors detect position change when an actuator moves the robotic arm, providing an analog voltage signal, proportional to the amount of motion, to a controller that makes appropriate adjustments based on programmable set points. With the data provided by the sensor, the controller can determine when the robotic arm should stop or slow down. For example, on a pick and place machine, if the robotic arm exceeded its travel distance, it could drive right through the board it intended to populate. Macro Sensors miniature linear position sensors evaluate the robotic arm movement for optimum performance.
In addition to industrial robots, Macro Sensors CD 375 miniature LVDTs are suitable as an integral part of devices, such as hydraulic actuators, servo valves, medical equipment and other small mechanisms. A compact 3/8” diameter design and lightweight low mass core make the small contactless position sensors ideal for applications having high dynamic response requirements such as ATMs, copy machines, plastic injection molding machines and automatic inspection equipment.
Available in full-scale measurement ranges from ±0.025 inch to ±2.0 inch, these ac operated linear position sensors offer a core-to-bore radial clearance of 0.012 inch. Units feature repeatability (error of <0.01% of FSO), high resolution and the high sensitivity consistent with good linearity (maximum linearity error of ±0.25% FRO).
Environmentally robust, the CD 375 Series operate in temperature extremes of -65 °F to 220 °F. Stainless steel housings and epoxy encapsulation protects units from environmentally hostile conditions. External sealing of the miniature position sensors meets IEC standard IP61.
Interchangeable with most manufacturer units, the CD 375 Series LVDTs operate with any conventional differential input LVDT signal conditioners. Macro Sensors manufactures a full line of LVDT signal conditioners that offer optimum performance from any CD 375 Series LVDTs.
- Edited by Gust Gianos, Control Engineering, www.controleng.com
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
2012 Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.