Calibrating multi-variable flowmeters
If you’re using more than one variable from a single device, you need to calibrate all of them sooner or later. Video: Beamex demonstrates dual pressure calibration.
One of the advantages of modern process sensor technology is that you can often measure more than one type of process variable with only one device and one penetration into the process. Probably the most common example of this is using a differential pressure (DP) sensor with an orifice plate or other device to measure flow. In most situations, the sensor can also give you a pressure reading of the line and probably a process temperature as well.
If you’re using all those readings, your quality assurance program will typically require you to calibrate all those functions as part of your larger sensor calibration routines. Some field calibration devices, such as the Beamex MC6, can handle the differential pressure, static pressure, and temperature devices in one operation.
In the attached video, Ned Espy discusses how this approach works, and demonstrates how it can be done on a Rosemount 3051S multi-variable transmitter. The test tools used in this case include two pressure gages, RTD simulator, mA meter, and HART communicator. The methods used for testing can be applied to calibration and verification of other flowmeters and complex smart instruments.
Peter Welander, pwelander(at)cfemedia.com
Case Study Database
Get more exposure for your case study by uploading it to the Plant Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.
Annual Salary Survey
In a year when manufacturing continued to lead the economic rebound, it makes sense that plant manager bonuses rebounded. Plant Engineering’s annual Salary Survey shows both wages and bonuses rose in 2012 after a retreat the year before.
Average salary across all job titles for plant floor management rose 3.5% to $95,446, and bonus compensation jumped to $15,162, a 4.2% increase from the 2010 level and double the 2011 total, which showed a sharp drop in bonus.